Description
Overview
With advanced microprocessor technology and state-of-the-art silicon pressure sensors, the Additel 160 series Intelligent Digital Pressure Module (IDPM) provides an accurate, reliable, and economic solution for wide range of pressure applications. It is loaded with functionality and remarkably easy to use. In order to reach the best performance, every silicon pressure sensor in the module has been specially aged, tested and screened before assembly. Designed as external pressure module for Additel 760 automatic handheld pressure calibrator, Additel 761 automated pressure calibrator and Additel 222A, Additel 223A and Additel 761 calibrators, the Additel 160 IDPM is unmatched in performance and reliability.
Features
• Pressure ranges to 60,000 psi (4,200 bar)
• Precision accuracy to 0.01% RD
• Pressure measurement accuracy of 0.02% FS
• Fully temperature compensated accuracy
Applications
The 160A series Digital Pressure Modules (IDPM) are specially designed as external pressure module for 761 automated pressure calibrator, 222A Multifunction Process Calibrators and 223A Documenting Process Calibrators.
Specifications
Standard Accuracy | Precision Accuracy | ||||
---|---|---|---|---|---|
CPXXX | GPXXX | DPXXX | APXXQ | APXXR | |
Operating temperature | -10°C to 50°C (14°F to 122°F) | 15°C to 25°C (59°F to 77°F) | 10°C to 30°C (50°F to 86°F) | ||
Storage temperature | -20°C to 70°C (-4°F to 158°F) | -20°C to 70°C (-4°F to 158°F) | |||
Relative humidity | 95% RH | 90% RH | |||
Pressure connections (for external use only) | 1/4NPT, 1/2NPT, 1/4BSP, 1/2BPS, M20x15 | Festo 6mm | 1/4NPT, 1/2NPT, 1/4BSP, 1/2BPS, M20x15 | ||
Enclosure (for external use only) | SS enclosure | ||||
Dimensions (Dia x H) | 33 mm x 123 mm (1.3” x 4.84”) | Depends on the model | 44 mm x 195 mm (1.7” x7.7”) | ||
Weight | 0.4 kg (0.99 lb) | Depends on the model | 0.7 kg (1.5 lb) | ||
Warranty | 1 Year | 1 Year | 1 Year |
Gauge Pressure[1] | |||||
---|---|---|---|---|---|
P/N | Pressure Range | Media[2] | Accuracy (%FS) | Burst Pressure | |
(inH2O) | (mbar) | ||||
V15 | -15 | -1 | G | 0.02 | 3× |
GP2 | 2 | 0.16 | G | 0.05 | 3× |
GP5 | 5 | 0.35 | G | 0.05 | 3× |
GP10 | 10 | 0.7 | G | 0.02 | 3× |
GP15 | 15 | 1 | G | 0.02 | 3× |
GP30 | 30 | 2 | G | 0.02 | 3× |
GP50 | 50 | 3.5 | G , L | 0.02 | 3× |
GP100 | 100 | 7 | G , L | 0.02 | 3× |
GP150 | 150 | 10 | G , L | 0.02 | 3× |
GP300 | 300 | 20 | G , L | 0.02 | 3× |
GP500 | 500 | 35 | G , L | 0.02 | 3× |
GP600 | 600 | 40 | G , L | 0.02 | 3× |
GP1K | 1000 | 70 | G , L | 0.02 | 3× |
GP2K | 2000 | 140 | G , L | 0.02 | 3× |
GP3K | 3000 | 200 | G , L | 0.02 | 3× |
GP5K | 5000 | 350 | G , L | 0.02 | 3× |
GP10K | 10000 | 700 | G , L | 0.02 | 2× |
GP15K | 15000 | 1000 | G , L | 0.05 | 2x |
GP20K | 20000 | 1400 | G , L | 0.05 | 1.5x |
GP25K | 25000 | 1600 | G , L | 0.1 | 1.5x |
GP30K | 30000 | 2000 | G , L | 0.1 | 1.5x |
GP36K | 36000 | 2500 | G , L | 0.1 | 1.5x |
GP40K | 40000 | 2800 | G , L | 0.1 | 1.35x |
GP50K | 50000 | 3500 | G , L | 0.1 | 1.2x |
GP60K | 60000 | 4200 | G , L | 0.1 | 1.1x |
[1]. Sealed gauge pressure for above 1000 psi [2]. G=Gas, L=Liquid |
Differential Pressure | ||||||
---|---|---|---|---|---|---|
P/N | Pressure Range[1] | Media | Accuracy (%FS) | Burst Pressure | Static Pressure Range | |
(inH2O) | (mbar) | |||||
DP1 | ±1 | ±2.5 | G | 0.05[2] | 100× | ±10 psi |
DP2 | ±2 | ±5.0 | G | 0.05[2] | 100× | ±10 psi |
DP5 | ±5 | ±10 | G | 0.05[2] | 50× | ±10 psi |
DP10 | ±10 | ±25 | G | 0.05[2] | 20× | ±10 psi |
DP20 | ±20 | ±50 | G | 0.05 | 20× | ±10 psi |
DP30 | ±30 | ±75 | G | 0.05 | 20× | ±10 psi |
DP50 | ±50 | ±160 | G | 0.05 | 3× | ±10 psi |
DP150 | ±150 | ±350 | G | 0.02 | 3× | 50 psi |
DP300 | ±300 | ±700 | G | 0.02 | 3× | 50 psi |
[1] FS specification applies to the span of the range. Accuracy includes 1 year stability. [2] 0.05%FS accuracy (incl 6 months stability). One year accuracy is 0.05%FS calibration accuracy combined with 0.05%FS one year stability. |
Compound Pressure | ||||||
---|---|---|---|---|---|---|
P/N | Pressure Range | Media | Accuracy | Pressure Rating | ||
psig | bar.g | Burst | Over Pressure | |||
CP2 | ±2 | ±0.16 | G | 0.05% FS | 3x | 1.2x |
CP5 | ±5 | ±0.35 | G | 0.02% FS | 3x | 1.2x |
CP10 | ±10 | ±0.7 | G | 0.02% FS | 3x | 1.2x |
CP15 | ±15 | ±1.0 | G | 0.02% FS | 3x | 1.2x |
CP30 | -15 to 30 | -1 to 2.0 | G | 0.02% FS | 3x | 1.2x |
CP50 | -15 to 50 | -1 to 3.5 | G | 0.02% FS | 3x | 1.2x |
CP100 | -15 to 100 | -1 to 7.0 | G,L | 0.02% FS | 3x | 1.2x |
CP300 | -15 to 300 | -1 to 20 | G,L | 0.02% FS | 3x | 1.2x |
CP500 | -15 to 500 | -1 to 35 | G,L | 0.02% FS | 3x | 1.2x |
CP600 | -15 to 600 | -1 to 40 | G,L | 0.02% FS | 3x | 1.2x |
CP1K | -15 to 1,000 | -1 to 70 | G,L | 0.02% FS | 3x | 1.2x |
CP2K | -15 to 2,000 | -1 to 140 | G,L | 0.02% FS | 3x | 1.2x |
CP3K | -15 to 3,000 | -1 to 200 | G,L | 0.02% FS | 3x | 1.2x |
CP5K | -15 to 5,000 | -1 to 350 | G,L | 0.02% FS | 3x | 1.2x |
CP10K | -15 to 10,000 | -1 to 700 | G,L | 0.02% FS | 2x | 1.2x |
Absolute Pressure | |||||
---|---|---|---|---|---|
P/N | Pressure Range | Media | Accuracy (%FS) | Burst Pressure | |
psi | bar | ||||
AP5 | 5 | 0.35 | G | 0.1 | 3× |
AP10 | 10 | 0.7 | G | 0.1 | 3× |
AP15 | 15 | 1 | G | 0.1 | 3× |
AP30 | 30 | 2 | G | 0.1 | 3× |
AP50 | 50 | 3.5 | G | 0.1 | 3× |
AP100 | 100 | 7 | G , L | 0.05 (0.1) | 3× |
AP300 | 300 | 20 | G , L | 0.05 (0.1) | 3× |
AP500 | 500 | 35 | G , L | 0.05 (0.1) | 3× |
AP1K | 1000 | 70 | G , L | 0.05 (0.1) | 3× |
AP3K | 3000 | 200 | G , L | 0.05 (0.1) | 3× |
AP5K | 5000 | 350 | G , L | 0.05 (0.1) | 3× |
Precision Sensors[1] – Absolute Pressure | ||||||
---|---|---|---|---|---|---|
P/N | Pressure Range | Media | Accuracy[2] | Pressure Rating | ||
psia | bar.a | Burst | Over Pressure | |||
AP15R | 0 to 15 | 0 to 1 | G | 0.01% FS | 2x | 1.2x |
AP30R | 0 to 30 | 0 to 2 | G | 0.01% rdg or 0.003% FS whichever is greater | 2x | 1.2x |
AP50R | 0 to 50 | 0 to 3.5 | G | 0.01% rdg or 0.003% FS whichever is greater | 2x | 1.2x |
AP100R | 0 to 115 | 0 to 8 | G,L | 0.01% rdg or 0.003% FS whichever is greater | 2x | 1.2x |
AP300R | 0 to 315 | 0 to 21 | G,L | 0.01% rdg or 0.003% FS whichever is greater | 2x | 1.2x |
AP500R | 0 to 515 | 0 to 36 | G,L | 0.01% rdg or 0.003% FS whichever is greater | 2x | 1.2x |
AP1KR | 0 to 1,015 | 0 to 71 | G,L | 0.01% rdg or 0.003% FS whichever is greater | 2x | 1.2x |
AP2KQ | 0 to 2,015 | 0 to 141 | G | 0.005% rdg + 0.005% FS | 3x | 1.1x |
AP3KQ | 0 to 3,015 | 0 to 201 | G | 0.005% rdg + 0.005% FS | 3x | 1.1x |
AP6KQ | 0 to 6,000 | 0 to 400 | G,L | 0.01% FS | 3x | 1.2x |
AP10KQ | 0 to 10,000 | 0 to 700 | G,L | 0.01% FS | 2x | 1.2x |
AP15KQ | 0 to 15,000 | 0 to 1000 | G,L | 0.01% FS | 2x | 1.2x |
AP20KQ | 0 to 20,000 | 0 to 1400 | G,L | 0.01% FS | 2x | 1.2x |
AP30KQ | 0 to 30,000 | 0 to 2000 | G,L | 0.02% FS | 1.5x | 1.1x |
AP40KQ | 0 to 40,000 | 0 to 2800 | G,L | 0.02% FS | 1.5x | 1.1x |
[1] Contact Additel for other range options. [2] Accuracy includes calibration uncertainty, linearity and long-term stability. |
Barometric Pressure | |||||
---|---|---|---|---|---|
P/N | Pressure Range | Media | Accuracy | Burst Pressure | |
Low | High | ||||
BP | 60 kPa | 110 kPa | G | 40 Pa | 3× |
Reviews
There are no reviews yet.